Novel Corona Virus (SARS-CoV-2) Ag Rapid Test Kit
Manufactured by Jiangsu Bioperfectus Technologies Co. Ltd., China - https://www.bioperfectus.com/
Device identification number
HSC common list (RAT)
Cassette, Lateral flow
Nasal swab, Nasopharyngeal swab, Oropharyngeal swab
Cross-reactivity (pathogens tested)
B.1.1.7 (Alpha), B.1.351 (Beta), B.1.429 (Epsilon), B.1.617.1 (Kappa), B.1.617.2 (Delta), B.1.617.3, B.1.621 (Mu), C.37 (Lambda), P.1 (Gamma)
2022-08-24 09:08:01 CET
Analysis of cross reactivity
Past medical history and other diagnostic information is necessary to determine infection status.
Negative results should be treated as presumptive and confirmed with other diagnostic techniques.
Type of antigen
The database contains publicly available In Vitro Diagnostic Medical Devices for COVID-19 and it is being updated periodically. Please note that additional performance (as retrieved from manufacturers web pages) is provided only for devices commercially available with CE-IVD mark. Acknowledgements