DIAQUICK COVID-19 Ag Cassette
Manufactured by DIALAB GmbH, Austria -
Device identification number
HSC common list (RAT)
Cross-reactivity (pathogens tested)
B.1.1.529 (Omicron), B.1.1.7 (Alpha), B.1.351 (Beta), B.1.427 (Epsilon), B.1.525 (Eta), B.1.526 (Iota), B.1.617.1 (Kappa), B.1.617.2 (Delta), C.37 (Lambda), P.1 (Gamma), P.2 (Zeta), P.3 (Theta)
2022-08-24 09:08:01 CET
It is a twin test of Alltest (ID 1257). The tests are identical in design and construction, but distributed under a different name.
2022-02-03 Removed from list
Analysis of cross reactivity
0.34 % (total 2023 negative samples. 7 wrong positive. 2016 correct negative.)
1.98 % (total 101 positive samples. 99 correct positive. 2 wrong negative)
93.4 % (relative sensitivity)
99.9 % (relative specificity)
Type of antigen
The database contains publicly available In Vitro Diagnostic Medical Devices for COVID-19 and it is being updated periodically. Please note that additional performance (as retrieved from manufacturers web pages) is provided only for devices commercially available with CE-IVD mark.